- •The intrarater reliability and ecological validity of a novel and true-to-form jar instrument was tested in women with hand osteoarthritis
- •Intrarater reliability of the jar instrument was good to excellent and participants reported the experience of using it to be highly representative of jar-opening at home.
- •The jar tool may be used in future study of the required hand kinetics for successful opening of a sealed jar and for testing of joint protection strategies in this population.
Jar opening is a task that relies heavily on the ability to generate adequate hand forces and is often reported in the literature as being difficult for women with hand arthritis. Many have studied relationships between diminished grip/pinch strength and occupational performance but few have investigated how much hand force is necessary to successfully engage in manual tasks. Those who studied this have relied on approaches and tools which lack ecology. Additionally, few have instrumented daily objects to understand if joint protection techniques do, in fact, reduce the hand force generation when performing manual tasks.
Purpose of the Study
The objectives of this study were to one) determine the within session repeatability of a device used to quantify these forces and two) probe into the ecological validity of a novel device to be used in the future study of women with hand osteoarthritis to measure hand force requirements and study the impact of joint protection interventions on hand force profiles.
A plastic jar was instrumented with a torque limiter, 6-axis load cell, and six force sensing resistors so as to capture the grip and compressive hand forces which act on a jar lid when opening a sealed jar. To assess intra-rater reliability of the tool and its testing procedures, 31 adult women with hand osteoarthritis were asked untwist the jar's lid so as to break its seal with each hand twice while stabilizing the base with the opposing hand. The agreement between trials of peak forces and torques from each hand was assessed through statistical approaches including Intraclass Correlation Coefficient, Standard Error of the Measurement, and Minimal Detectable Change. The jar's ecological validity was then assessed via survey.
This instrument and methods yielded good to excellent repeatability across all force outputs. The majority of our subjects (87%) reported the jar to be similar to those used at home, 87% reported to use a similar jar 2-3 times/month or greater, and rated the importance of opening such a jar as being, on average, 8.7/10.
The jar instrument appears to have high reliability and ecologic validity. It has the potential to reveal hand force requirements for a population known to have difficulties opening jars and understanding these force thresholds could help to inform therapy goal-setting. Beyond this, it has the potential to support the study of the hand forces used across various joint protection approaches so as to inform best “jar-opening” practices.
This tool was designed to represent the qualities of a standard, large sealed jar while still housing sophisticated kinetic measurement capacities. Our findings support that we have presented a tool which can be used in future study within this population to better understand the hand kinetics associated with the highly problematic task of jar-opening and joint protection strategies intended to reduce hand loads.
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Journal of Hand Therapy
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- Prevalence and most common causes of disability among adults–United States, 2005.Morb Mortal Wkly Rep. 2009; 58: 421-426
- Symptomatic hand osteoarthritis in the United States: prevalence and functional impairment estimates from the third U.S. national health and nutrition examination survey, 1991-1994.Am J Phys Med Rehabil. 2007; 86: 12-21https://doi.org/10.1097/phm.0b013e31802ba28e
- Estimates of the prevalence of arthritis and other rheumatic conditions in the United States. Part II.Arthritis Rheum. 2008; 58: 26-35https://doi.org/10.1002/art.23176
- Dimensionality and clinical importance of pain and disability in hand osteoarthritis: Development of the Australian/Canadian (AUSCAN) Osteoarthritis Hand Index.Osteoarthr Cart. 2002; 10: 855-862https://doi.org/10.1053/joca.2002.0837
- Activity limitations and participation restrictions in women with hand osteoarthritis: patients’ descriptions and associations between dimensions of functioning.Ann Rheum Dis. 2005; 64: 1633-1638https://doi.org/10.1136/ard.2004.034900
- Canadian Occupational Performance Measure.5th (ed.). CAOT Publications, Ottowa2014
- Clinical Features.in: Altman RD Lozada CJ Rheumatology. Mosby, New York2003
- OARSI-FDA initiative: defining the disease state of osteoarthritis.Osteoarthr Cart. 2011; 19: 478-482https://doi.org/10.1016/j.joca.2010.09.013
- Relationship of radiographic and clinical variables to pinch and grip strength among individuals with osteoarthritis.Arthritis Rheum. 2005; 52: 1424-1430https://doi.org/10.1002/art.21035
- Radiological assessment of osteo-arthrosis.Ann Rheum Dis. 1957; 16: 494-502https://doi.org/10.1136/ard.16.4.494
- The effect of hand osteoarthritis on grip and pinch strength and hand function in postmenopausal women.Clin Rheumatol. 2003; 22: 420-424https://doi.org/10.1007/s10067-003-0792-4
- Evaluation of individual finger forces during activities of daily living in healthy individuals and those with hand arthritis.J Hand Ther. 2020; https://doi.org/10.1016/j.jht.2020.04.002
- The rheumatoid thumb.in: Mackin E Callahan A Skirven T Al. E Rehabilitation of the Hand: Surgery and Therapy. 3rd (ed). Mosby, St. Louis1990: 929
- The handling of consumer packaging.Appl Ergon. 1981; 12: 153-161https://doi.org/10.1016/0003-6870(81)90004-1
- Modelling wrist-twisting strength of the elderly.Ergonomics. 1988; 31: 1807-1819https://doi.org/10.1080/00140138808966829
- Grip strengths and required forces in accessing everyday containers in a normal population.Am J Occup Ther. 1998; 52: 621-626https://doi.org/10.5014/ajot.52.8.621
- The relationship between hand strength and the forces used to access containers by well elderly persons.Am J Occup Ther. 2002; 56: 78-85https://doi.org/10.5014/ajot.56.1.78
- A force transducer to measure individual finger loads during activities of daily living.J Biomech. 1999; 32: 721-725https://doi.org/10.1016/s0021-9290(99)00054-8
- The effect of jar holding posture on finger force and torque during a jar-opening task for young females.Packag Technol Sci. 2014; 27: 265-276https://doi.org/10.1002/pts.2031
- Jar-opening challenges. Part 1: An apparatus for assessing hand and finger torques and forces in a jar-opening activity.Proc Inst Mech Eng Part H J Eng Med. 2009; 223: 121-129https://doi.org/10.1243/09544119JEIM422
- A novel device for evaluating forces during the dynamic jar opening action-Do older and younger adults do things differently?.Med Eng Phys. 2011; 33: 521-525https://doi.org/10.1016/j.medengphy.2010.11.017
- Distribution of grip force in three different functional prehension patterns.J Med Eng Technol. 2006; 30: 176-182https://doi.org/10.1080/03091900600565217
- Evaluation of handles in a maximum gripping task.Ergonomics. 2004; 47: 1350-1364https://doi.org/10.1080/00140130410001715697
- Development, construct validity, and reproducibility of a mimetic sealed jar measuring the dynamics of opening.J Rehabil Assist Technol Eng. 2017; 4https://doi.org/10.1177/2055668317692222
- Anthropometry and strength relationship: male-female differences.Anthropol Anzeiger. 1998; 56: 49-56
- Physician diagnosed arthritis, reported arthritis and radiological non-axial osteoarthritis.Osteoarthr Cart. 2008; 16: 846-850https://doi.org/10.1016/j.joca.2007.12.001
- The interaction between lid diameter, height and shape on wrist torque exertion in younger and older adults.Ergonomics. 2002; 45: 922-933https://doi.org/10.1080/00140130210162243
AMTI. FS6-100 specifications. https://amti.biz/Products/FS/FS6.aspx. Published 2020. Accessed March 31, 2020.
- Fundamentals of Packaging Technology.3rd (ed). Institute of Packaging Professionals, Naperville2002
- The AIMS2-SF: a short form of the arthritis impact measurement scales 2. french quality of life in rheumatology group.Arthritis Rheum. 1997; 40: 1267-1274https://doi.org/10.1002/1529-0131(199707)40:7>1267::AID-ART11<3.0.CO;2-L
- Short-form Arthritis Impact Measurement Scales 2: tests of reliability and validity among patients with osteoarthritis.Arthritis care Res. 1999; 12: 163-171https://doi.org/10.1002/1529-0131(199906)12:3>163::aid-art3<3.0.co;2-z
- The prehensile movements of the human hand.J bone Jt Surg - Br. 1956; 38-B (Vol): 902-913
- Foundations of Clinical Research: Applications to Practice.3rd. Pearson/Prentice Hall, Saddle River2009
- Guidelines, criteria, and rules of thumb for evaluating normed and standardized assessment instruments in psychology.Psychol Assess. 1994; 6: 284-290https://doi.org/10.1037/1040-35126.96.36.1994
- Sample size and optimal designs for reliability studies.Stat Med. 1998; 17: 101-110
- Thumb base involvement in symptomatic hand osteoarthritis is associated with more pain and functional disability.Ann Rheum Dis a J Clin Rheumatol Connect tissue Res. 2010; 69: 585-587https://doi.org/10.1136/ard.2009.104562
- The case for using the repeatability coefficient when calculating test-retest reliability.PloS one. 2013; 8: e73990https://doi.org/10.1371/journal.pone.0073990
JHT Read for Credit
Quiz: # 875
Record your answers on the Return Answer Form found on the tear-out coupon at the back of this issue or to complete online and use a credit card, go to JHTReadforCredit.com. There is only one best answer for each question.
- # 1.One stated purpose of the study was to determine the ___________ of the jar instrument
- # 2.Participating subjects were
- a.patients with either OA, or RA of the hand
- b.all normal student volunteers
- c.post op CMC arthroplasty patients
- d.either physician-diagnosed, or exhibiting symptoms of hand OA
- # 3.The most significant expression of the required force was
- a.measured in ft lbs
- b.rotational force
- c.opening torque
- d.opening power
- # 4.Participants felt the jar device was
- a.too difficult to maneuver repeatedly
- b.reasonably similar to the activity they experience at home
- c.too easy to open
- d.uncomfortable in actual use
- # 5.The authors conclude that the device is suitable for clinical application
When submitting to the HTCC for re-certification, please batch your JHT RFC certificates in groups of 3 or more to get full credit.
Published online: April 15, 2021
Accepted: April 5, 2021
Received in revised form: March 2, 2021
Received: April 16, 2020
© 2021 Elsevier Inc. All rights reserved.